Conductivity / Resistivity Controller Big Screen Hot Sales High Accuracy
High pure/ultrapure water measurement control, widely used for electronic, electric, pharmacy, fine chemistry, clinical medicine, life science.
● Widely used for monitoring of all water treatment process, which have high performances in high salt water treatment, concentrated solution, recirculating cooling water, reclaimed water and ect.
● High pure/ultrapure water measurement control, widely used for electronic, electric, pharmacy, fine chemistry, clinical medicine, life science.
Con: 0～3500μS/cm; (C=0.1cm-1)
TDS: 0～17500ppm; (C=1.0cm-1)
Res: 0～18.25MΩ·cm; (C=0.05cm-1)
|Temp.Comp.||The 25 ℃ basis, automatic temperature compensation|
|Oper. Temp.||Con/TDS: -25℃～+ 125℃; Res：0～100℃|
JL-1.0-PT1K; JF-1.0-PT1K; JL-10.0-PT1K
|Display||2x16 Bit LCD|
|Current output||Isolation could be relocated 4～20mA|
|Control output||Programmable: High limit or low limit relay|
the maximum pulse rate: 400 Pulses/Min
RS485; Baud rate: 1200, 2400, 4800
Ambient temperature: 0～50℃
Relative humidity: ≤85%
|Installation mode||Panel mounted (embedded)|
1.High pure water and ultrapure water could not select open sampling measurement method. If the high pure water is exposed to the air, a lot of carbon dioxide would dissolve into the water immediately and the impurity inside the container and dusts in the air would also dissolve into the water which could cause serious errors. So high pure water could only adopt inclosed, running and side flow circulation method. With open measurement method to test high pure water by using laboratory appliance is inadvisable.
2. Resin regeneration dirts would pollute the cell, please clean by alcohol cotton ball.